Vacuum Flow Rate Guide
- Conveniently check your own flow rates without the need for an engineer
- Ensure compliance with BS EN ISO 10637
BS EN ISO 10637
Current international standards which ensure the safety of dental suction systems require the following:
Suction systems require a minimum of 250 litres of air per minute at a 16mm (plus or minus 1mm) aspirator tip with a minimum vacuum of 140 mbar and a maximum vacuum of 250 mbar.
To read our full blog post about vacuum flow rates and why they are important click here.
How to use the gauge (PDF)
Payment & Security
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.